This standard was last reviewed and confirmed in 2015. Therefore this version remains current.
Silicon content can be determined in the range of 0,2 % (m/m) to 1 % (m/m). The principle is dissolution of a test portion in acid, aspiration of the test solution into a nitrous oxide-acetylene flame of an atomic absorption spectrometer, measurement of the absorbance of the resonance line energy from the spectrum of silicon and comparison with that of calibration solutions at 251,6 nm.
Status : PublishedPublication date : 1992-09
Edition : 1Number of pages : 3
Technical Committee:Nickel and nickel alloys
Buy this standard
A standard is reviewed every 5 years
Revisions / Corrigenda